Cleaning activator for the alkaline oxidative cleaning of medical instruments by the OXIVARIO™ and OXIVARIO PLUS™ method. Can be combined with Sekumatic™ FR and Sekumatic™ MultiClean.
Cleaning activator for alkaline cleaning.
|Our performance||Your advantage|
|Oxidative cleaning intensifier||Excellent oxidative cleaning support against tough protein stains together with alkaline cleaning agents|
|Optimally matched formulation components||Compatible with the alkaline instrument cleaners Sekumatic™ FR and Sekumatic™ MultiClean|
|Secured method||Can be used in the OXIVARIO™ and OXIVARIO PLUS™ method. Extensively tested and released by the MIELE company|
In cleaning and disinfection devices in the medical area. Usable for all water hardness grades
The addition is made in the 2nd Cleaning phase of the Oxivario™ method with a dosing of 3.5ml/l Sekumatic™ Oxivario™ to 5ml/l Sekumatic™ FR ore Sekumatic™ MultiClean and in the Oxivario™ Plus method, with 7ml/l Sekumatic™ Oxivario™ to 8ml/l Sekumatic™ FR or Sekumatic™ MultiClean.
In the OrthoVario method, the addition is done in the 2nd Cleaning step with a dosing of 4ml/l to 3ml/l Sekumatic™ NeutraClean and 1.5ml/l Sekumatic™ FR.
In the subsequent rinsing phase after the 2nd Cleaning step, alkaline residues are neutralized with Sekumatic™ FNZ in a dosing of 1-3ml/l.
The Sekumatic™ Oxivario™/ Sekumatic™ FR / Sekumatic™ MultiClean solution must be completely rinsed off after the contact. For final rinsing and thermal disinfection, use fully demineralized water (DI).
Please follow the information provided by the instrument manufacturer.
Only for professional use
Hydrogen peroxide solution; surfactant-free
|In container1||Until the printed expiration date|
Please note the following to ensure the correct storage of environmental hygiene and instrument disinfectants: The original containers (canisters, buckets or bottles) with the disinfectant concentrate (or powder) should be stored at temperatures below 25°C and kept dry and clean. A brief period below or above the given storage temperature (e.g. during transport) will not diminish the effectiveness. The containers are to be sealed tightly after each use. Please also take care that the concentrate remaining in the container is not contaminated when preparing the application solution.
|Chemical & Physical Data||Temperature||Value|
|Density||20 °C||1.11 g / ml|
|pH value* (concentrate)||20 °C||approx. 3|
|pH-value* in OXIVARIO™ method (solution, 5 ml / L Sekumatic™ FR ** + 3.5ml/l Sekumatic™ OXIVARIO™ in DI water)||20 °C||approx. 11|
|pH-value* in OXIVARIO™ Plus method (solution of 8ml/l Sekumatic™ FR ** + 7ml/l Sekumatic™ OXIVARIO™ in DI water)||20 °C||approx. 11|
* pH-value measurements under operating conditions always by using pH-meter with temperature compensation
** Alternative Sekumatic® MultiClean
Please see below related offerings:
Our search returned no results.